Assembly of extracorporeal circuit with Efferon® device and hemofilter in “POSTfilter” mode on PrismaFlex RRT machine
0,9% saline solution bag, 1 liter
Plastic bag for priming solution,
Assemble the extracorporeal circuit of the RRT machine
according to its user manual.
Add 10.000 ME heparin to saline solution and mix it to prepare the rinsing solution.
Connect DIN-Luer bloodline 1 with priming solution bag. Fill the bloodline with the priming solution using gravity. Attach the clamp to the bllodline.
Mount the Efferon® device in vertical position.** .
Remove the cap from bottom port of the Efferon® device and connect the pre-filled DIN-Luer bloodline 1 to it.
Remove the cap from upper port of the Efferon® device and connect the empty DIN-Luer bloodline 2 to it.
Connect the plastic bag for rinsing solution to DIN-Luer bloodline 2.
Remove the clamp from DIN-Luer bloodline 1 and rinse the Efferon® device using gravity
Put the clamps to DIN-Luer bloodlines 1 and 2 after the priming is finished
Disconnect the plastic bags from DIN-Luer bloodlines 1 and 2 and utilize it
Connect the Luer-Luer
adapter to DIN-Luer bloodline 2
Stop the blood pump
of RRT machine
Put the clamps on inlet
and outlet bloodlines of the hemofilter
Disconnect the extracorporeal
circuit between the hemofilter
outlet and air bubble trap.
Сonnect the bloodlinesin a sequence: DIN-Luer bloodline 1 is attached to the hemofilter, and Luer-Luer adapter is attached to the air bubble trap.
The circuit is ready. Remove all the clamps and start the blood pump.
the extracorporeal treatment:
— 160-210 seconds of ACT
— 60-80 seconds of aPTT